You are viewing the weighing news Stable Performance of General Measure’s Weighing Indicators Tested under OIML Requests at Golden Lotus Measuring equipment.
General Measure Technology Co. Ltd. (China) – Press Release: To make sure all the weighing indicators work stable in harsh environments, General Measure has tested the weighing indicators by high and low-temperature test, anti-static test, anti-burst test, and anti-lighting surge test in our R&D laboratory. It turns out GM’s products have superior performance.
1. Indicator Low-Temperature Test
Bạn mong muốn tìm kiếm một cân công nghiệp chính hãng chất lượng cao hay một cân điện tử giá rẻ, hãy cho Hoa sen vàng biết và nhận sự tư vấn nhiệt tình-chu đáo, sản phẩm đều có giá trị và đặc tính riêng của chúng … hãy Gọi ngay số : 028 3511 7799 hoặc truy cập website: https://hoasenvang.com.vn/shops/ (2021)
Demonstration: After calibrating the weighing indicator, we put it in temp&humi chamber and set the temperature as -40℃. After 3 hours in low temp, we came back to check the working status of the indicator by turning the simulator. The weighing indicator still works as normal with a quick response.
2. Indicator Anti-Static Test
Demonstration: In this test, we prepared the weighing indicator GMT-P1, weighing scale with 30kg full-scale loadcell, and static discharging gun. After connecting the weighing indicator with loadcell and calibrating with 200g weight, we used a static discharging gun to do contact discharging to the metal case of the weighing indicator, then air discharging to the non-metal case, by following the OIML R76 requirements as ±6KV and ±8KV voltage respectively. Mostly, the weighing indicator remains 200g, which is very stable.
3. Indicator Anti-Burst Test
Demonstration: Regarding the anti-burst test, we utilized a fast transient burst simulator and coupling clamp to test the signal end of the weighing indicator. Under the OIML R76 requirement, we set voltage as ±0.5KV to positive and negative pole interfere in 120s respectively. During the test, the weighing indicator kept displaying 200g all the time, which is very stable.
4. Indicator Anti Lightning Surge Test
Demonstration: Talking about anti lightning surge test, the lightning surge generator was connected to the weighing indicator that is connected to the loadcell. In the real experiment, we set voltage as 1KV that is higher than the requirement of OIML R76 (0.5KV), but the result turns out that the weighing indicator is very stable without any display variation.
Through the results from the above 4 tests, we are very proud to introduce our stable weighing indicators to the clients and industry.
✨The latest news at Blog.hoasenvang.com.vn, news on precision Measuring Equipment and the weighing control devices, Stable Performance of General Measure’s Weighing Indicators Tested under OIML Requests all free news and non-profit, Yes.. All purpose serving you is my pleasure.
🎁 [High quality Ads] Can dien tu cong nghiep chinh xac GLS at Cân siêu thị Hoa sen vàng – Golden Lotus Measuring Equipment, weighing scales news, electronic scale news, load cell news, retail scale news, industrial scale news, … International Electronic Weighing for Feature Thanks to Weighing_review internet site for World comprehensive Weighing Scales News. All noesis on the Blog is doled out and is not accountable for the noesis doled out for any commercial message intention or inexactness data. Please do not check, all changes are successful to the producer without prior detect.